Eliminating the risk of inadvertent connection to medical air via a flowmeter
A National Patient Safety Alert has been issued asking all providers that use piped medical air to eliminate the risk of inadvertently connecting patients to medical air via a flowmeter instead of oxygen.
- National Patient Safety Alert – Eliminating the risk of inadvertent connection to medical air via a flowmeter
About this alert
Air flowmeters attached to piped medical air outlets are primarily used to drive the administration of nebulised medication; typically for short periods to manage respiratory conditions. Most other uses of piped medical air do not require an air flowmeter.
Due to the proximity of the piped medical air and oxygen outlets at the bedside, and the similarity in design of flowmeters, there is a significant risk when using air flowmeters that patients may be inadvertently connected to medical air instead of oxygen.
The alert asks providers to purchase alternative devices that do not require medical air to be delivered via an air flowmeter. Following this, all medical air flowmeters except those tethered to equipment for niche use should be discarded, and all medical air outlets no longer required should be reversibly capped off.
About National Patient Safety Alerts
This alert has been issued as a National Patient Safety Alert.
The NHS England and NHS Improvement patient safety team was the first national body to have been accredited to issue National Patient Safety Alerts by the National Patient Safety Alerting Committee (NaPSAC). All National Patient Safety Alerts are required to meet NaPSAC’s thresholds and standards. These thresholds and standards include working with patients, frontline staff and experts to ensure alerts provide clear, effective actions for safety-critical issues.
NaPSAC requires providers to introduce new systems for planning and coordinating the actions required by any National Patient Safety Alert across their organisation, with executive oversight.
Failure to take the actions required under any National Patient Safety Alert may lead to CQC taking regulatory action.
Latest News from
NHS England proposes new mental health access standards22/07/2021 12:20:00
The NHS is set to take another major step towards improving patient access to mental health services with the introduction of five new waiting time guarantees, under plans set out today.
NHS England announces new Innovative Medicines Fund to fast-track promising new drugs21/07/2021 12:20:00
NHS patients are set to benefit from early access to potentially life-saving new medicines, including cutting-edge gene therapies, thanks to a new Innovative Medicines Fund and £680 million of ringfenced funding, the NHS chief executive has announced today.
NHS workforce more diverse than any point in its history, as health service commits to more action on representation20/07/2021 16:10:00
A first of its kind report looking into race equality among England’s doctors has found that the number from black and ethnic minority backgrounds working for the NHS is the highest on record.
NHS ‘Grab A Jab’ Weekend In Final Drive To Get The Country Protected16/07/2021 16:43:00
People can ‘grab-a-jab’ when they pop into Primark or while sunbathing in the park as the NHS vaccination programme seeks to maximise protection before Covid restrictions are lifted.
NHS Patients, staff and visitors must continue to wear face coverings in healthcare settings16/07/2021 13:48:00
England’s Chief Nurse yesterday reminded the public that everyone accessing or visiting healthcare settings must continue to wear a face covering and follow social distancing rules.
NHS text message drive to encourage millions to bring forward second vaccine16/07/2021 09:15:00
Hundreds of thousands of text messages will be sent by the NHS COVID-19 Vaccination programme this week reminding people to bring forward their second dose appointment.
NHS investment boost to ambulance staff numbers ahead of winter15/07/2021 16:10:00
Ambulance trusts in England will be given an extra £55 million to boost staff numbers ahead of winter, the NHS recently (14 July 2021) announced.
Inappropriate anticoagulation of patients with a mechanical heart valve15/07/2021 09:15:00
A National Patient Safety Alert has been issued on the risk of inappropriate anticoagulation of patients with a mechanical heart valve.